Vapor Phase Hydrogen Peroxide (VPHP) Generator
HYPER Model – Your reliable partner in 6 log bio-decontamination processes
HYPER is a Vapor Phase Hydrogen Peroxide (VPHP) generator specifically designed to perform bio-decontamination processes of clean, dry, sealed enclosures used in medical, pharmaceutical, and biological applications.
HYPER works in single pass (open loop) configuration to achieve very fast bio-decontamination cycles.
HYPER is controlled by a Siemens PLC S7-1200 to ensure a safe control process and generate a complete bio-decontamination cycle report.
Communication with the isolator’s control system via PROFINET or discrete I/O.
Depending on your configuration, HYPER can be either integrated into the isolator system or remoted in technical areas.
To minimize operators’ exposure to the liquid hydrogen peroxide during handling of bulk product, HYPER uses specially designed disposable containers containing 3,000 ml of 35% Hydrogen Peroxide (H₂O₂).
The bio-decontamination process is effective against a wide range of microorganisms. Processes can be validated with Geobacillus stearothermophilus biological indicators to routinely achieve 6-log sporicidal reduction.
The bio-decontamination process is a dry process resulting in no condensation of VPHP onto treated surfaces.
HYPER® provides a low temperature bio-decontamination method in which an aqueous solution of 35% H₂O₂ is flash vapourized into a (pre)heated air stream which enters the enclosure enriched with VPHP.
The cycle can be divided into four main phases:
Dry, High Efficiency Particulate Air (HEPA) filtered air is delivered to the enclosure to lower the relative humidity down to a predetermined level.
This phase permits the necessary target VPHP concentration to be maintained below saturation (dew point) levels during the Rising and Plateau phases.
The flow of dry HEPA filtered air continues while H₂O₂ is injected into the air stream.
The injection rate is controllable in the 0.0 to 30.0 grams per minute range, allowing you to reach the target VPHP concentration in the fastest way possible.
Rising time is affected by injection rate, airflow rate, enclosure volume, enclosure contents, and temperature.
The enclosure is constantly furnished with a VPHP enriched air volume from the generator, while the exhaust system extracts the same volume to a “safe area” or to an external VPHP catalysing system. The catalytic converter degrades the VPHP antimicrobial into oxygen and water vapour. The target VPHP sterilant concentration, reached at the end of the rising phase, is maintained for a specific time throughout the enclosure. Plateau time is affected by injection rate, airflow rate, enclosure volume, enclosure contents, and temperature.
VPHP injection is stopped and the flow of dry HEPA filtered air is delivered to the enclosure to reduce the concentration of H₂O₂ to a predetermined level.
Once such level is reached, the ventilation system of the enclosure completes the reduction of the VPHP concentration to a safe level (≤1ppm) by drawing a maximum air volume from the installation area and expelling the same volume to a “safe area” or to an external VPHP catalysing system.